LEVel for the value being entered. The front panel keys arefunctional. This disables all frontpanel keys except the Local key. After you call these VIs, the instrument is ready to take measurements or to stimulate a system. The dc source has three groups of statusregisters; Operation, Questionable, and Standard Event. Action VIs are different from Configuration VIs because they do not change the instrument settings but order the instrument to carry out an action based on its current configuration.
|Date Added:||23 August 2011|
|File Size:||57.87 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
After the instrument has been configured, the Read VI is used to read multiple points.
DISPlayThis command turns the front panel display on or agllent. It is intended to beused at the end of a command line so that the application program can then monitor the bus for data untilit receives the “1” from the dc source Output Queue.
The dc source has three groups of statusregisters; Operation, Questionable, and Standard Event. Commands that affect triggeractions are among the parallel commands.
Keysight Technologies / Agilent Technologies hp66xxbc Power Supply
CURRentAgilent A, A OnlyThis command defines a band around the trigger level through which the signal must pass before aninternal measurement can occur. Reprints of this guide containing minorcorrections and updates may have the same printing date. When all of these settings are set and the VI labviwe executed the results will be displayed in the Measurements array indicator. Generally, you need to call the Close Labbiew only once at the end of an application or when you finish communication with an instrument.
Page 31 Programming the DC Source – ! Page laview Programming the DC Source – 3Trigger occurs on rising edgeTrigger occurs on falling edgewhen signal crosses positivewhen signal crosses negativehysteresis band limithysteresis band limitTRIG: I am using a NA Power Supply.
How to Use an Instrument Driver in LabVIEW Tutorial – National Instruments
The calibration mode must be enabled before thewill accept any other calibration commands. It is also recommended that you set Disable Auto Serial Polling.
Display commands labivew the front panel display of the. NEGativetriggering occurs on the falling edge. Parallel commands allow othercommands to begin executing while the parallel command is still executing.
Common commands also perform system functions.
Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. The Utility VIs- perform a variety of operations that are auxiliary to the most often used instrument driver VIs.
The second parameter is the password. December 5, Page 61 Language Dictionary – 4Status CommandsStatus commands program 6632h dc source status registers. In this example, serial communication is used. Asshown in figurethe group consists of the same type of registers as the Status Operation group. Or point us to the URL where the manual is located.
Chapter 1Introduction to this guide. The output mode switch is used to connect ordisconnect the the output capacitor located inside the unit.
What’s missing? Tell us about it.